Receive your quote directly from the manufacturer. Organization: Asante Three Rivers Medical Center Ease of use 5 out of 5 After sales service 5 out of 5 Value for money 3 out of 5 Easy to use analyzer with small sample volume. Instrument use cost is high. Rating: 4. Alifax ESR analysers are the only ones capable to provide results in 20 seconds by measuring red blood cells aggregation overcoming the variables and limitations of the sedimentation method. Manufacturer Alifax S.
|Country:||United Arab Emirates|
|Published (Last):||2 January 2005|
|PDF File Size:||10.31 Mb|
|ePub File Size:||2.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Download this manual See also: Service Manual. In Vitro Diagnostic Medical Device for professional use. All rights reserved. Unauthorized copying of this manual or parts of it is prohibited. Table of Contents. Related Manuals for Alifax Roller 20 No related manuals.
Page 4 We reserve the right to make changes in the course of technical development without previous notice. Neither this manual nor any parts of it may be duplicated or transmitted in any way without the written approval of Alifax S. Available for 1, Ord. Handling of Instructions for use Manual the instructions for use manual is provided for your safety and gives important instructions for the handling of the system described.
Close the door to allow the system to progress with the analysis. Damaged connecting cables must be replaced immediately! During the analysis the instrument will display on the screen the result obtained. Considering the instrument uses a Teflon tube in which blood, water and latex flows, it is normal that the internal walls of the capillary tend to become opaque, and also to remain dirty because some blood residual parts remain inside the capillary.
After have removed the tests-tube from positions 1 and 2, the instrument, independently from its configuration, requires the insertion of an ulterior test-tube always with distilled water in position 1. This is useful because all the hydraulic circuitry remains filled with water or with detergent CBC detergent.
The procedure for calibration is : 1. This function becomes useful if a haematology mixer is not available during the comparative proofs between the instrument and the method used in laboratory. The diagram visualizes the trend of washing values detected by the three sensors, which are directly correlated to the photometric signal. This is particularly significant with increasing the samples considered daily. The average ESR and Page 3. The Latex Controls kit is a valid tool for the Ord.
The starting reference are the values obtained during the calibration and then, for every control executed with the Latex Kit, the trend will show how much the read latex values drifts from the reference values printed on the box and also on the printout report generated during each control procedure. With this option it is possible to visualize precise information about all Page 1.
As explained before, numbers are typed using the keyboard. In the Flag List are printed operational parameters of the instrument. Date and time of the CPS Board software compilation. As is possible see, the tech level password allow to access the previously explained functions SW version, Print in run, Print exp, Wash time, Reset Maint plus others which will be explained in the next pages.
This counter increases every time an analysis is executed. When the threshold of maintenance is reached up by default is set to , the instrument will begin to advise the necessity of technical maintenance.
In this mode, instrument rotor mix the sample with a special function, called centrifugation, that is much powerful and fast than normal mixing. Unscrew the upper side of the piston as shown in the photo.
This operation has to be done in safety. Discard the replaced needle. Take the new needle, remove the transparent cap that protects the needle thread. More than one carriage position sensor detected active at the same time physically impossible! Technical service repair or check inside the instrument. Protection tools and suggested materials to be used: Glasses. Remember that an excessive tension of belt enhances the motor noise level during rotation.
X Version 1. This document MUST be attached to the instrument. Print page 1 Print document pages. Cancel Delete. Sign In OR. Don't have an account? Sign up! Restore password. Upload from disk. Upload from URL.
Roller 20 PN
Alifax Roller 20 User Manual